AGILENT 33220A MANUAL PDF
The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Agilent A. 20 MHz Function /. Arbitrary Waveform Generator. Service Guide . Publication Number (order as manual set). Agilent A. Function/Arbitrary Waveform Generator .. trigger from the rear- panel Trig In connector, a manual trigger from the key, or a software (bus) trigger .
|Published (Last):||8 July 2012|
|PDF File Size:||4.86 Mb|
|ePub File Size:||16.69 Mb|
|Price:||Free* [*Free Regsitration Required]|
MIN selects the lowest frequency allowed for the selected function and MAX selects the highest frequency allowed. When agklent used as an input, the Trig In connector can be configured as an output to enable the A to trigger other instruments at the same time as its internal trigger occurs.
Page Chapter 5 Error Messages Execution Errors Settings conflict; not able to sweep noise, sweep turned off The mwnual generator cannot generate a sweep using the noise function. The commands used for AM are listed on the next page. For the A, you do not have to change the length of the waveform to change its output frequency. Press the Hop Freq softkey and then set 33220a value to Hz using the numeric keypad or the knob and cursor keys.
A list of help topics is also available to assist you with several front-panel operations. At this point, the function generator outputs a continuous sweep from 50 Hz to 5 kHz if the output is enabled. The factory default is Chapter 5 Error Messages Command Errors Program mnemonic too long A command header was received which contained more than the maximum 12 characters allowed. Page Chapter 3 Features and Functions Calibration Overview To Change the Security Code To change the security code, you must first unsecure the function generator, and then enter a new code.
Page Index softkey labels 4 questionable data register trigger output software bus trigger,register diagram trigger source standard event register sweeps software revision status byte register Sync signal software, connectivity 15 status system symmetry 65 sound beeper This section is only an overview and does not give all of the details you will need to write your own application programs. MIN selects the most negative dc offset voltage for the function and amplitude specified.
Chapter 3 Features and Functions Pulse Waveforms Pulse Waveforms A shown below, a pulse waveform consists of a period, a pulse width, a rising edge, and a falling edge. You can name a location from the front panel or over the remote interface, but you can only recall a state by name from the front panel. The default is 1 ms. The security code is stored in non-volatile memory.
You cannot use pulse, noise, or dc as the carrier waveform.
If a command requires agielnt than one parameter, you must separate adjacent parameters using a comma as shown below: Used only when the Internal modulation source is selected PM: Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the sweep mode, the function generator outputs a single sweep when a trigger signal is received. The default is 0 degrees.
In AM, the amplitude of the carrier is varied by the instantaneous voltage of the modulating waveform.
MAX selects the largest dc offset for the selected function and agileent. Chapter 3 Features and Functions Output Configuration Duty Cycle Square Waves The duty cycle of a square wave represents the amount of time per cycle that the square wave is at a high level assuming that the waveform is not inverted.
The rear-panel Trig connector cannot be used for both operations at the same time.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Page Chapter 5 Error Messages Command Errors Program mnemonic too long A command header was received which contained more than the maximum 12 characters allowed. The default is 1 kHz for all functions. Page Chapter 3 Features and Functions Frequency Sweep Marker Frequency If desired, you can set the frequency at which the signal on the front- panel Sync connector goes to a logic low during the sweep.
All of the standard output functions except dc and noise have an associated Sync signal. You can output the burst at a rate determined by the internal rate generator or the signal level on the rear-panel 332220a In connector.
Press any of the front-panel function keys except. Page Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds for the sweep. Chapter 7 Tutorial Pulse Waveform Generation Pulse Waveform Generation To eliminate distortion due to aliasing at higher frequencies, the Agilent A also uses a different waveform generation technique to create pulse waveforms.
While in the waveform editor, you define the waveform by specifying time and voltage values for each point in the waveform. The following steps show you how to change the amplitude to 50 mVrms. The subnet mask indicates the portion of the host address to be agjlent to indicate the subnet.
The sweep mode has been turned off.
Agilent Technologies 33220A User Manual
Page Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the sweep mode, the function generator outputs a single sweep when a trigger signal is received. The default is no password. Page 89 Chapter 3 Features and Functions Phase Modulation PM Phase Deviation The phase deviation setting represents the peak variation in phase of the modulated waveform from the carrier waveform.
Select an internal or external FSK source. Its combination of bench-top and system features makes this function generator a versatile solution for your testing requirements now and in the future. The default is NORM true-high logic. Chapter 5 Error Messages Execution Errors Data out of range; large period limits minimum pulse width For a pulse waveform, the minimum pulse width is 20 ns for periods up to 10 seconds.
The default is Hz for all functions. You can select a sine, square, ramp, pulse, or arbitrary waveform noise is allowed only in the gated burst mode and dc is not allowed. Burst Mode Overview The following is an overview of the steps required to generate a burst. If you want to modify the example programs, or write your own programs and compile them, you will need to install the Agilent IO Libraries Suite software. Page Index FREQuency command center frequencyfront-panel operation 38 FUNCtion command external trigger source FSK rate 39, 91 modulating frequency 82, frequency spanhop frequency 38, 91, modulating source 84, front-panel operation 40, 42 modulating source 92, modulating waveform 82 linear vs.
If you set the carrier to a value less than the deviation frequency with FM enabledthe function generator will automatically adjust the deviation to the maximum value allowed with the present carrier frequency.
Used in the internal triggered burst mode only.