HEMOCHRON RESPONSE PDF
HEMOCHRON Response Coagulation Monitoring Instrument from ITC. Used and refurbished Hemochron Response Coagulation Analyzer from Soma Technology. Request a quote online or call Description. The HEMOCHRON® Response is a whole blood coagulation monitoring instrument designed to be used at the point-of-care. The Response.
|Published (Last):||25 February 2017|
|PDF File Size:||7.91 Mb|
|ePub File Size:||3.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
Verify that the quality control materials and test tubes are he,ochron expiration date. This item will be shipped through the Global Shipping Program and includes international tracking.
Hemochron Response Brochure – Accriva Diagnostics – PDF Catalogs | Technical Documentation
The Hemochron Response is a portable, battery operated instrument designed to perform whole blood coagulation tests. And routine testing can continue. Holding the tube vertically, flick the tube 4 to 6 times. Cooling fan is replaced as needed.
Calibrated to OEM specifications. For CA test tubes: Any international shipping is paid in part to Pitney Bowes Inc. When clot formation is detected, the instrument beeps three times and the test name, clotting time in seconds is displayed.
Repeat the test for the second well. Quality control results and maintenance is reviewed weekly by a Medical Technologist. Transfer the 2 cc into the Hemochron test tube.
It is reusable and has no expiration date. Watch list is full. Peel back the flip top on the distilled, deionized water vial, but do not remove the crimp seal use this technique when opening each heomchron. Instrumentation Laboratory Inova Diagnostics.
HEMOCHRON Response Coagulation Monitoring Instrument from ITC
Special financing available Select PayPal Credit at checkout to have the option to pay over time. Before drawing the blood sample, hold test tube vertically and tap the tube bottom several times on a horizontal surface to shake activator particles down into testing zone.
Do not obtain blood from heparinized access line, lock or indwelling heparin lock. Use a backup instrument until problem is resolved see page 2 for list of instruments and locations.
Contact the seller – opens in a new window or tab and request a shipping method to your location. Flip open the cap of the Hemochron plastic test tube and dispense 0.
However, serine protease inhibitors such as aprotinin that may be administered to certain patients to decrease postoperative bleeding can prolong the celite activated ACT. Please enter a number less than or equal to 9. Room temperature storage is possible for non-punctured vials; room temperature dating tesponse to a maximum of seven weeks, but must never exceed the marked expiration date. The results may be interpreted by the clinician as an indication of the status of anticoagulant therapy at the time of sample collection.
Other offers may also be available.
Observance of incompetency will necessitate re-inservicing and inability to use analyzer until competency is acceptable. Since the magnet has been displaced, it is no longer sensed by the instrument’s magnetic detector. Interpretation of Test Results: Werfen in Colombia Phone: The instrument gives an audible beep and displays the coagulation time.
A beep will signal the start of the test, and timing of the test begins. The Hemochron ACT tests are affected by poor technique, including specimen collection and test procedure. This can be performed by any operator or a Medical Technologist.
If the instrument terminates a test in this manner, a FAULT message is displayed indicating the test result is outside of the specified range. Reesponse increase depends upon how large the dose is and the duration of the procedure.